- 1 Synopsis
- 2 Location
- 3 Versioning
- 4 Functional Description
- 5 Integration into BCI2000
- 6 Block Diagram
- 7 Parameters
- 8 State Variables
- 9 See also
An environment extension which manages multichannel, low latency audio I/O.
Griffin Milsap (firstname.lastname@example.org)
06/11/2012: Initial public release;
Source Code Revisions
- Initial development: 4095
- Tested under: 4095
- Known to compile under: 4095
- Broken since: --
- Fix Known Issues
- Add per-sample resolution to envelopes
- Leaving the module running for long periods of time in halted state causes a long time of no state logging before signal goes to realtime. Seems to be unrelated to how long system was left running (~12-15 seconds) -- Not sure if this is an issue with the extension itself, or an issue with the bcievent interface.
- Bandpass filtering in filterbanks doesn't appear to function
Experiments which require audio input or real-time audio synthesis based on system state are now possible with the AudioExtension. This extension is capable of recording multiple channels of audio input, synthesizing tones or noise, and reading encoded audio files. These channels are input to a mixing matrix which mixes those inputs to multiple channels of audio output. Both input and output are run through a simple filterbank, then they have their envelope extracted and logged into states via the bcievent interface. Audio input and output channels can be recorded into audio files losslessly and can be resynchronized offline. The mixing matrix is a matrix of expressions which can be used to dynamically change audio mixing based on the system state.
Integration into BCI2000
Compile the extension into your source module by enabling contributed extensions in your CMake configuration. You can do this by going into your root build folder and deleting
CMakeCache.txt and re-running the project batch file, or by running
cmake -i and enabling BUILD_AUDIOEXTENSION. Once the extension is built into the source module, enable it by starting the source module with the
--EnableAudioExtension=1 command line argument.
The eyetracker is configured in the Source tab within the EyetrackerLogger section. The configurable parameters are:
LogEyetracker- Enables/Disables logging of Eyetracker states
NetworkLocation- The network address of the Eyetracker given by the Tobii Eyetracker Browser
Port- The port that the Tobii communicates over - Tobii default is 4455
LogGazeData- Enables/Disables logging of gaze data
LogEyePos- Enables/Disables logging of eye position (as seen from the camera)
LogPupilSize- Enables/Disables logging of pupil size (very rough)
LogEyeDist- Enables/Disables logging of the distance from the screen to the eyes (again, rough)
GazeScale- Scales the incoming gaze data first
GazeOffset- Offsets the incoming gaze data after scaling
Note: GazeScale and GazeOffset are quick hacks to address an issue with gaze data being clamped around the edges of the screen. The eyetracker gives back values which are between 0.0 and 1.0 for onscreen gaze but supports looking slightly offscreen by allowing gaze data returned to go above 1.0 and below 0.0. BCI2000 needs this scaled between 0.0 and 1.0 before the gaze data is multiplied by 65535 for storage in the 16 bit state. These two parameters account for this scaling and offset and prevents the clamping from happening as often as it would otherwise. These parameters will be removed once BCI2000 supports typed states.
The following code retreives the actual ~(0.0-1.0) range that the eyetracker outputs directly (assuming you've scaled and offset the signal to avoid clipping) from each eye and averages it to find a gaze position.
float x = State( "EyetrackerLeftEyeGazeX" ) + State( "EyetrackerRightEyeGazeX" ); x /= ( 2.0f * 65535.0f ); float y = State( "EyetrackerLeftEyeGazeY" ) + State( "EyetrackerRightEyeGazeY" ); y /= ( 2.0f * 65535.0f ); x -= ( float )Parameter( "GazeOffset" ); x /= ( float )Parameter( "GazeScale" ); y -= ( float )Parameter( "GazeOffset" ); y /= ( float )Parameter( "GazeScale" );
Unless otherwise specified, all states are prefixed with
Eyetracker<Left/Right>Eye which corresponds with each individual eye. The EyetrackerLogger extension does not support subjects with more than two eyes at the moment.
The eye gaze position (where - on the screen - the subject is looking) is returned from the Tobii SDK as 32 bit floating point numbers which (roughly) range from 0.0 to 1.0. They are multiplied by 65535 and stored as 16 bit integers in these states if the
LogGazeData parameter is enabled. (0,0) corresponds to the top left of the screen, (65535,65535) corresponds to the right bottom of the screen. -- See EyetrackerStatesOK.
The eye position relative to the camera in 2D space is returned if
LogEyePos is enabled. Again, these are returned from the library as floating point numbers from 0.0 to 1.0 and are scaled to 16 bit integer values from 0 to 65535. (0,0) corresponds to the top left of the camera's view, and (65535,65535) corresponds to the bottom right of the camera's view.
The pupil size in mm is saved in this state if
LogPupilSize is enabled. It corresponds to the length of the longest chord drawn from one side of the pupil to the other. The size will change depending on the eye position and distance from the screen. Although it is given in mm, it would be best to use this as a relative measurement.
The distance between the screen and the eyes in mm is saved in this state if
LogEyeDist is enabled. This measurement is an approximation. The actual measurement will depend on whether or not the test subject is wearing glasses or not.
This state is a number from 0 to 4 and is documented in the Tobii SDK manual. It is repeated here for convenience.
- 0 - The eye tracker is certain that the data for this eye is right. There is no risk of confusing data from the other eye.
- 1 - The eye tracker has only recorded one eye and made some assumptions and estimations regarding which is the left and which is the right eye. However, it is still very likely that the assumption made is correct. The validity code for the other eye is in this case always set to 3.
- 2 - The eye tracker has only recorded one eye, and has no way of determining which one is the left eye and which one is the right eye. The validity code for both eyes is set to 2.
- 3 - The eye tracker is fairly confident that the actual gaze data belongs to the other eye. The other eye will always have validity code 1.
- 4 - The actual gaze data is missing or definitely belonging to the other eye.
|Code (Right - Left)||Description|
|0 - 0||Both eyes found. Data is valid for both eyes.|
|0 - 4 or 4 - 0||One eye found. Gaze data is the same for both eyes.|
|1 - 3 or 3 - 1||One eye found. Gaze data is the same for both eyes.|
|2 - 2||One eye found. Gaze data is the same for both eyes.|
|4 - 4||No eye found. Gaze data for both eyes are invalid.|
It'd probably be wise to remove all data points with a validity state of 2 or higher while running your analysis.
Early versions of the extension didn't take into account that the library may return a number greater than 1.0 or less than 0.0. This resulted in "pac-man" style wrap around of gaze coordinates in 2.0 and crashes in 3.0. If the output from the library is out of bounds, it is clamped to the boundaries and the "EyetrackerStatesOK" parameter is changed. A value of "1" corresponds to valid gaze data, a value of "0" corresponds to invalid "clamped" gaze data. Use the "GazeOffset" and "GazeScale" parameters to avoid clamping. Those parameters scale and offset the data so that when it does go out of range, it can still be fit into the 16 bit state.